Rotor and Tube Holders Included
HORIZON 6 Flex
Programmable Routine Centrifuge
Max RCF: 2,000 xg
Max RPM: 3,800 RPM
Weight: 12 lbs (5.4 kg)
Dimensions: 9 x 14 x 12 in (23 x 36 x 30 cm)
Warranty: 2 years (lifetime on rotor)
- Monitor time remaining on the digital display
- Program up to 10 custom time and speed settings
- LED lid lighting indicates cycle status
- Designed, built, and supported in the USA
Ready To Ship
Easy to Use
Easy to Monitor
Process samples more efficiently. HORIZON's LED lid lighting is off when the centrifuge is ready to load, on when running, and flashing at the end of each cycle.
Engineered for Safety
Video Tutorial Library
Specifications & Presets
- Height – 9 in. (23 cm)
- Width – 12 in. (30 cm)
- Depth – 14 in. (36 cm)
- Weight – 12 lbs. (5.4 kg)
- Sound Level – 64 dBA
- Max. Acceleration Time – 20 Seconds
- Max. Deceleration Time – 20 Seconds
- Factory Setting Range – 1 to 30 Minutes (± 1%)
- Voltage – 95 to 253 VAC
- Frequency – 50/60 Hz
- Max. Power Requirements – 220 Watts
- Max Speed – 3,800 RPM
Preset 1: 3,600 RPM | 10 minutes | 1,800 xg
Preset 2: 3,300 RPM | 15 minutes | 1,500 xg
Preset 3: 1,900 RPM | 5 minutes | 500 xg
Manuals & Literature
What is the warranty on Drucker's centrifuges?
Drucker Diagnostics provides a two (2) year parts and labor warranty (except the rotor - see question below).
What is the warranty on the rotor?
All rotors carry a lifetime warranty. Should the rotor ever fail, for as long as you own your centrifuge, Drucker Diagnostics will replace it free of charge.
Will I need to purchase accessories for the centrifuge separately?
All of our centrifuges are sold complete with the rotor and tube holders. No additional accessories are necessary though there are additional options available for purchase.
How often should my centrifuge be calibrated?
Drucker Diagnostics recommends the centrifuge be calibrated every 2 years or 5,000 cycles.
What should be used to clean and disinfect the HORIZON 6 Flex?
ONLY isopropyl alcohol, soap and water, or a 10% (5,500 PPM) bleach solution should be used for cleaning and disinfection of the centrifuge and accessories. All surfaces must be dried immediately after cleaning and disinfecting.
What cleaning solutions should be avoided?
TBQ Germicidal products should not be used as they will cause damage to the centrifuge and void the warranty. The use of fully/partially halogenated hydrocarbons, ketones, esters, ethers, benzyls, ethyl benzenes, and all other chemicals not prescribed by Drucker Diagnostics should not be used as they may cause damage to the rotor chamber, rotor, tube holders, accessories, and centrifuge exterior. If this occurs the warranty will be voided.